The distance moved by the screw of a screw gauge is
in four rotations and there are
divisions on its cap. When nothing is put between its jaws,
divisions of circular scale coincides with reference line, and zero of linear scale is hidden from circular scale when two jaws touch each other or zero circular scale is laying above the reference line. When plate is placed between the jaws, main scale reads
divisions and circular scale reads
divisions. Thickness of plate is.